async def _do_tp(self, pip, mount) ->top_types.Point:
    """ Execute the work of tip probe.

        This is a separate function so that it can be encapsulated in
        a context manager that ensures the state of the pipette tip tracking
        is reset properly. It should not be called outside of
        :py:meth:`locate_tip_probe_center`.

        :param pip: The pipette to use
        :type pip: opentrons.hardware_control.pipette.Pipette
        :param mount: The mount on which the pipette is attached
        :type mount: opentrons.types.Mount
        """
    pip.update_instrument_offset(top_types.Point())
    hotspots = robot_configs.calculate_tip_probe_hotspots(pip.
        current_tip_length, self._config.tip_probe)
    new_pos: Dict[Axis, List[float]] = {ax: [] for ax in Axis.gantry_axes() if
        ax != Axis.A}
    safe_z = (self._config.tip_probe.z_clearance.crossover + self._config.
        tip_probe.center[2])
    for hs in hotspots:
        ax_en = Axis[hs.axis.upper()]
        overridden_center = {ax: (sum(vals) / len(vals) if len(vals) == 2 else
            self._config.tip_probe.center[ax.value]) for ax, vals in
            new_pos.items()}
        x0 = overridden_center[Axis.X] + hs.x_start_offs
        y0 = overridden_center[Axis.Y] + hs.y_start_offs
        z0 = hs.z_start_abs
        pos = await self.current_position(mount)
        await self.move_to(mount, top_types.Point(pos[Axis.X], pos[Axis.Y],
            safe_z))
        await self.move_to(mount, top_types.Point(x0, y0, safe_z))
        await self.move_to(mount, top_types.Point(x0, y0, z0))
        if ax_en == Axis.Z:
            to_probe = Axis.by_mount(mount)
        else:
            to_probe = ax_en
        async with self._motion_lock:
            self._current_position = self._deck_from_smoothie(self._backend
                .probe(to_probe.name.lower(), hs.probe_distance))
        xyz = await self.gantry_position(mount)
        self._log.debug(
            'tip probe: hs {}: start: ({} {} {}) status {} will add {}'.
            format(hs, x0, y0, z0, new_pos, xyz[ax_en.value]))
        new_pos[ax_en].append(xyz[ax_en.value])
        bounce = self._config.tip_probe.bounce_distance * (-1.0 if hs.
            probe_distance > 0 else 1.0)
        await self.move_rel(mount, top_types.Point(**{hs.axis: bounce}))
        await self.move_to(mount, xyz._replace(z=safe_z))
    to_ret = top_types.Point(**{ax.name.lower(): (sum(vals) / len(vals)) for
        ax, vals in new_pos.items()})
    self._log.info(
        'Tip probe complete with {} {} on {}. New position: {} (default {}), averaged from {}'
        .format(pip.name, pip.pipette_id, mount.name, to_ret, self._config.
        tip_probe.center, new_pos))
    return to_ret

Given the source code of a python function, suggest a fitting name for the function.
system
async def _do_tp(self, pip, mount) ->top_types.Point:
    """ Execute the work of tip probe.

        This is a separate function so that it can be encapsulated in
        a context manager that ensures the state of the pipette tip tracking
        is reset properly. It should not be called outside of
        :py:meth:`locate_tip_probe_center`.

        :param pip: The pipette to use
        :type pip: opentrons.hardware_control.pipette.Pipette
        :param mount: The mount on which the pipette is attached
        :type mount: opentrons.types.Mount
        """
    pip.update_instrument_offset(top_types.Point())
    hotspots = robot_configs.calculate_tip_probe_hotspots(pip.
        current_tip_length, self._config.tip_probe)
    new_pos: Dict[Axis, List[float]] = {ax: [] for ax in Axis.gantry_axes() if
        ax != Axis.A}
    safe_z = (self._config.tip_probe.z_clearance.crossover + self._config.
        tip_probe.center[2])
    for hs in hotspots:
        ax_en = Axis[hs.axis.upper()]
        overridden_center = {ax: (sum(vals) / len(vals) if len(vals) == 2 else
            self._config.tip_probe.center[ax.value]) for ax, vals in
            new_pos.items()}
        x0 = overridden_center[Axis.X] + hs.x_start_offs
        y0 = overridden_center[Axis.Y] + hs.y_start_offs
        z0 = hs.z_start_abs
        pos = await self.current_position(mount)
        await self.move_to(mount, top_types.Point(pos[Axis.X], pos[Axis.Y],
            safe_z))
        await self.move_to(mount, top_types.Point(x0, y0, safe_z))
        await self.move_to(mount, top_types.Point(x0, y0, z0))
        if ax_en == Axis.Z:
            to_probe = Axis.by_mount(mount)
        else:
            to_probe = ax_en
        async with self._motion_lock:
            self._current_position = self._deck_from_smoothie(self._backend
                .probe(to_probe.name.lower(), hs.probe_distance))
        xyz = await self.gantry_position(mount)
        self._log.debug(
            'tip probe: hs {}: start: ({} {} {}) status {} will add {}'.
            format(hs, x0, y0, z0, new_pos, xyz[ax_en.value]))
        new_pos[ax_en].append(xyz[ax_en.value])
        bounce = self._config.tip_probe.bounce_distance * (-1.0 if hs.
            probe_distance > 0 else 1.0)
        await self.move_rel(mount, top_types.Point(**{hs.axis: bounce}))
        await self.move_to(mount, xyz._replace(z=safe_z))
    to_ret = top_types.Point(**{ax.name.lower(): (sum(vals) / len(vals)) for
        ax, vals in new_pos.items()})
    self._log.info(
        'Tip probe complete with {} {} on {}. New position: {} (default {}), averaged from {}'
        .format(pip.name, pip.pipette_id, mount.name, to_ret, self._config.
        tip_probe.center, new_pos))
    return to_ret

user
do_tp
assistant