def x(sp, fxpeaks, poly_degree_wfit, wv_master, wv_ini_search=None,
    wv_end_search=None, wvmin_useful=None, wvmax_useful=None, geometry=None,
    debugplot=0):
    if len(fxpeaks) <= poly_degree_wfit:
        print('>>> Warning: not enough lines to fit spectrum')
        return None
    naxis1 = sp.shape[0]
    wv_master_range = wv_master[-1] - wv_master[0]
    delta_wv_master_range = 0.2 * wv_master_range
    if wv_ini_search is None:
        wv_ini_search = wv_master[0] - delta_wv_master_range
    if wv_end_search is None:
        wv_end_search = wv_master[-1] + delta_wv_master_range
    xchannel = fxpeaks + 1.0
    list_of_wvfeatures = arccalibration(wv_master=wv_master, xpos_arc=
        xchannel, naxis1_arc=naxis1, crpix1=1.0, wv_ini_search=
        wv_ini_search, wv_end_search=wv_end_search, wvmin_useful=
        wvmin_useful, wvmax_useful=wvmax_useful, error_xpos_arc=3,
        times_sigma_r=3.0, frac_triplets_for_sum=0.5, times_sigma_theil_sen
        =10.0, poly_degree_wfit=poly_degree_wfit, times_sigma_polfilt=10.0,
        times_sigma_cook=10.0, times_sigma_inclusion=10.0, geometry=
        geometry, debugplot=debugplot)
    title = 'Wavelength calibration'
    solution_wv = fit_list_of_wvfeatures(list_of_wvfeatures=
        list_of_wvfeatures, naxis1_arc=naxis1, crpix1=1.0, poly_degree_wfit
        =poly_degree_wfit, weighted=False, plot_title=title, geometry=
        geometry, debugplot=debugplot)
    if abs(debugplot) % 10 != 0:
        xplot = np.arange(1, naxis1 + 1, dtype=float)
        ax = ximplotxy(xplot, sp, title=title, show=False, xlabel=
            'pixel (from 1 to NAXIS1)', ylabel='number of counts', geometry
            =geometry)
        ymin = sp.min()
        ymax = sp.max()
        dy = ymax - ymin
        ymin -= dy / 20.0
        ymax += dy / 20.0
        ax.set_ylim([ymin, ymax])
        for feature in solution_wv.features:
            xpos = feature.xpos
            reference = feature.reference
            ax.text(xpos, sp[int(xpos + 0.5) - 1], str(reference), fontsize
                =8, horizontalalignment='center')
        print('Plot with identified lines')
        pause_debugplot(12, pltshow=True)
    return solution_wv

Given the source code of a python function, suggest a fitting name for the function.
system
def x(sp, fxpeaks, poly_degree_wfit, wv_master, wv_ini_search=None,
    wv_end_search=None, wvmin_useful=None, wvmax_useful=None, geometry=None,
    debugplot=0):
    if len(fxpeaks) <= poly_degree_wfit:
        print('>>> Warning: not enough lines to fit spectrum')
        return None
    naxis1 = sp.shape[0]
    wv_master_range = wv_master[-1] - wv_master[0]
    delta_wv_master_range = 0.2 * wv_master_range
    if wv_ini_search is None:
        wv_ini_search = wv_master[0] - delta_wv_master_range
    if wv_end_search is None:
        wv_end_search = wv_master[-1] + delta_wv_master_range
    xchannel = fxpeaks + 1.0
    list_of_wvfeatures = arccalibration(wv_master=wv_master, xpos_arc=
        xchannel, naxis1_arc=naxis1, crpix1=1.0, wv_ini_search=
        wv_ini_search, wv_end_search=wv_end_search, wvmin_useful=
        wvmin_useful, wvmax_useful=wvmax_useful, error_xpos_arc=3,
        times_sigma_r=3.0, frac_triplets_for_sum=0.5, times_sigma_theil_sen
        =10.0, poly_degree_wfit=poly_degree_wfit, times_sigma_polfilt=10.0,
        times_sigma_cook=10.0, times_sigma_inclusion=10.0, geometry=
        geometry, debugplot=debugplot)
    title = 'Wavelength calibration'
    solution_wv = fit_list_of_wvfeatures(list_of_wvfeatures=
        list_of_wvfeatures, naxis1_arc=naxis1, crpix1=1.0, poly_degree_wfit
        =poly_degree_wfit, weighted=False, plot_title=title, geometry=
        geometry, debugplot=debugplot)
    if abs(debugplot) % 10 != 0:
        xplot = np.arange(1, naxis1 + 1, dtype=float)
        ax = ximplotxy(xplot, sp, title=title, show=False, xlabel=
            'pixel (from 1 to NAXIS1)', ylabel='number of counts', geometry
            =geometry)
        ymin = sp.min()
        ymax = sp.max()
        dy = ymax - ymin
        ymin -= dy / 20.0
        ymax += dy / 20.0
        ax.set_ylim([ymin, ymax])
        for feature in solution_wv.features:
            xpos = feature.xpos
            reference = feature.reference
            ax.text(xpos, sp[int(xpos + 0.5) - 1], str(reference), fontsize
                =8, horizontalalignment='center')
        print('Plot with identified lines')
        pause_debugplot(12, pltshow=True)
    return solution_wv

user
wvcal_spectrum
assistant